The FiggHealth Control Solution contains a known concentration of glucose.
The FiggHealth Blood Glucose Control Solution is for use with the FiggHealth smart blood glucose meters and FiggHealth Strips as a quality control check to verify that the meter and test strip are working together properly, and that the test is performing correctly.
For self-testing use only.
You should perform a quality control test:
Three levels of control solution are available. They are Control Solution 1 (level 1), Control Solution 2 (level 2) and Control Solution 3 (level 3). Control Solution 2 (level 2) is sufficient for most of self-testing needs. If you think your meter or strips may not be working correctly, you may also want to do level 1 or level 3 tests. If you want to purchase control solution kit, contact FiggHealth Customer Support: +1-800-391-2802, Or visited our website FiggHealth for help.
Control Solution 1 contains less than 0.1% glucose (active ingredient). Control Solution 2 contains less than 0.2% glucose (active ingredient). Control Solution 3 contains less than 0.3% glucose (active ingredient). Three levels control solution have preservatives in an aqueous based mixture.
If you want to purchase control solution kit, or want to know more information, contact FiggHealth Customer Support: +1-800-391-2802, Or visited our website Figg.Health for help.
1. Insert a new test strip. The meter turns on. Refer to your meter's User Manual details on how to record the result as a quality control test, and more details on operating the meter.
2. Shake the control solution bottle thoroughly.
3. Squeeze the control solution bottle gently. Discard the first drop. If the tip clogs, tap the tip gently on a clean, hard surface. Shake again, and then use.
4. Squeeze out a second small drop on a clean nonabsorbent surface. Touch the sample tip of the test strip to the control solution drop until the meter beeps.
5. Read the result from the meter display after a beep.
Control solution test results should be within the control range. The ranges for three levels (Level 1, Level 2 and level 3) are displayed on the test strip vial label (or on the foil pouch). For confirmation of results, Control Solution 1 tests should fall within the level 1 range, Control Solution 2 tests should fall within the level 2 range, and Control Solution 3 tests should fall within the level 3 range. If the test results are in the respective ranges, this means your Blood Glucose Monitoring System is working right and you are doing the procedure correctly. If the control solution tests results do not fall within the respective range:
After checking everything listed above, repeat the control solution test with a new test strip. If your results still fall outside the range indicated on the test strip vial label or on the foil pouch, your meter may not be working properly. DO NOT use the system to test blood. Contact Customer Support for help. For complete instructions, please refer to the User’s Manual included with your meter. If attempts to correct a problem fail, contact FiggHealth Customer Support: +1-800-391-2802, Or visited our website FiggHealth for help.